We hereby confirm that masks manufactured and distributed by Medicom do not contain nanoform graphene,
or any type of graphene.
Please rest assured that the quality and safety of all Medicom products is our primary concern.
On this page, you will find our key hints and tips to help make fit testing as fast, easy and accurate as possible. We have also included:
- A recording of an online training session
- A copy of the presentation as a video for you to watch at your leisure
- Links to view and download our qualitative and quantitative training videos, as well as our donning and doffing videos
Please feel free to browse these resources and use for your own training.
About the SafeMask Architect Pro™ Respirator Mask
Made in Canada by the medical mask experts, this medical grade respirator has been evaluated for use in healthcare settings according to recognized standards for biocompatibility, flammability and fluid resistance. It offers optimal respiratory protection by filtering out over 95% of solid airborne particles free of oil for up to 8-hours.
- Approved by Health Canada, the IRSST and the CNESST.
- Premium materials, innovative “duck bill” shape, adjustable nose piece and lightweight construction ensure optimal breathability.
- Unique single split comfort band holds mask securely in place.
- Tested in our laboratories, as well as by a recognized independent laboratory, to ensure quality and compliance.
- Available in 3 sizes; small, medium and large.
Hints & Tips on Fit Testing the Medicom SafeMask Architect Pro™respirator mask
The following tips will help ensure the most accurate results when performing fit testing of the Medicom SafeMask Architect Pro™ mask using either the qualitative or quantitative (Portacount) testing method.
In addition to the usual steps for testing – the wearer must be clean shaven, must not have eaten or smoked recently – he or she must closely follow the specified donning instructions, paying particular attention to ensuring that the Portacount test the probe is fitted in the centre of the top portion of the mask, between the nose and the chin to one side.
Ensure that the nose piece is curved gently before beginning donning.
Ensure that the nose piece does NOT get pinched into a “V”.
CORRECT MASK POSITION
Ensure that the mask is located just below the chin, approximately 1 cm behind the jawbone, and NOT all the way back to the throat.
Ensure that there is plenty of space at the front of the mask so that when in place, it is not near the lips.
Ensure that the top strap sits high on the head and, when possible, the lower strap is below the ears.
SEALING THE MASK
Once the mask is correctly positioned, the key areas for fitting are the nose piece and across the cheeks.
Using two fingers on either side of the nose, working away from the nose and across the cheek bones, push firmly every 2mm, moulding the mask to the cheeks.
Ensure that the nose piece does not create a “V” shape.
The wearer should press firmly along the top of the mask to ensure a secure fit.
With the Portacount test the probe should be fitted in the centre of the top portion of the mask between the nose and the chin to one side.
The Medicom SafeMask Architect Pro™ mask is extremely comfortable and breathable. It is designed to be worn for up to one continuous work session of 8 hours. Please see our videos for donning, doffing and fit testing instructions towards the bottom of this page.
Note: Some of the following demonstrations were filmed using the Kolmi FFP3 respirator, which is equivalent in construction to the Medicom SafeMask Architect Pro respirator. Kolmi is a Medicom brand sold across Europe.
Visit the ASSTSAS website to view self-training materials that provide fit testers with some tips to consider when performing qualitative and quantitative fit tests using Medicom Architect Pro RPDs.Visit ASSTSAS
Please watch this video to see a Medicom Fit Test expert show the best way to prepare and fit the Medicom SafeMask Architect ProTM mask.
Please watch this video of a Medicom Fit Test expert demonstrating the best way to remove the Medicom SafeMask Architect Pro™ mask.
For detailed directions on the Qualitative Fit Test Process with the Medicom SafeMask Architect Pro™ mask, please watch our dedicated video.
For detailed directions on the Quantitative or Portacount Fit Test Process with the Medicom SafeMask Architect Pro™ mask, please watch our dedicated video.
TDS – SafeMask® Architect Pro™ 95PFE-L2 Respirator – 202914, 203014, 203114
TDS – SafeMask® ArchitectTM Procedure Earloop Face Mask ASTM Level 3
TDS – SafeMask® ArchitectTM JR Procedure Earloop Face Mask
SafeMask® Architect Pro™ 95PFE-L2 Respirator Fitting Instructions
SafeMask® Architect Pro™ 95PFE-L2 Respirator
SafeMask® Architect Earloop Face Mask
SafeMask® Architect JR Pediatric Earloop Face Masks
SafeMask® Architect Pro™ 95PFE-L2 Respirator
Please complete the form below to stay informed on:
- New training materials
- New information and updates